Another useful feature of the CAL3K Calorimeters
The CAL3K calorimeter range has two main compensations:
- Firing Wire and Cotton Compensation
- Sulphur Compensation
Minor compensations are :
- Fibre Compensation (for food analysis)
- Moisture Compensation
- Gelatine Capsule Compensation
A brief explanation :
Cotton Compensation : Compensates for the firing energy and the cotton thread. Entry in degrees Celsius. Default : 0.04°C.
Sulphur Compensation : Compensates for the formation of sulphuric acid. Entry in Kj/g. Default : 0Kj/g. It is disabled during calibration and calibration check.
Fibre Compensation : Used during food/feed analysis. Entry in percent (of mass). Default 0%.
Moisture Compensation : Must be used when reporting in DRY form. Entry in percent (of mass). Default : 0%.
Gelatine Capsules : Used with powdery substances. Eliminates the spiking procedure. Entry in degrees Celsius. Is enabled in SYSTEM DEFINITION. Default is set to standard DDS Gelatine Capsules.