The keyboard command functions are restricted via a password. The most used parameters (functions) are accessible without restrictions. As an example: The Setting of Benzoic Acid (BA) calibration standard is password restricted. The default password is ‘DDS3K’. Your own password can be set via the PC software.
Not everyone is the same and not all laboratories operate on the same guidelines. We have tried to make the CAL3K as flexible as possible and in return introduced complexity. The unit is shipped with the "DDS3K" key password installed and activated.Learn More
BENZOIC ACID STANDARD VALUE
The BA (Benzoic Acid) standard is normally 26.454 MJ/Kg. Sometimes it is required to change the setting to the supplied certification standard. This can be done as follows: From a PC connected to any of the ports, From the Keyboard (Password required).Learn More
The CAL3K was designed to work as a standalone system. That means it does not need a computer to operate. This is achieved by loading the "DEFAULT" settings (which we do before shipping). The default settings are what we think average settings, and are good for all kinds of determinations.Learn More