WHAT IS SPIKING IN CALORIMETRY
What is spiking? If a sample does not ignite easily or not at all in a calorimeter, then the spiking method of ignition can be used to promote complete combustion of the sample. Only once the sample is completely burned can an accurate determination of the sample be ascertained. A combustion aid is added to the sample, like a Benzoic Acid Tablet, which will easily ignite, and in turn ignite the non-combustible sample. The heat produced by the combustion aid must be subtracted from the total energy release. The DDS Calorimeters range does this automatically. So, when do you need to use spiking?
As a principle, when a sample is fired in a bomb calorimeter, you are looking for total combustion of the sample for an accurate result. A simple visual inspection of the bomb inside and the crucible is in most cases sufficient. Getting to understand your sample, like coal, animal feed or ammunition, is key to discovering incomplete burning with a simple visual inspection. Should you find small solid bits of the sample still remaining in the bomb; then you possibly need to spike your sample.
Non-combustion is probably the easiest spiking problem to solve. Should your sample not combust, or partly combust, after a determination, then you will need to spike it. This can be achieved in multiple ways. Certified Benzoic Acid tablets have a consistent burning property and are ideal for spiking. Depending on your sample, Technical Oil can also be used as a spike or combustion aid. If your sample grain is small enough, you could even use Gelatine Capsules. It really depends on your sample consistency and laboratory preference. We have successfully fired wood chip with a moisture content of over 75% using Benzoic Acid and also Technical Oil.
RATE OF BURN
Some samples, like fine wood chips, burn extremely fast in a bomb calorimeter. So fast in fact that an explosion occurs, throwing out the rest of the sample from the crucible. So again you don't have total combustion, not because it can't burn. To solve this problem is to either compress the sample into a pellet or to use a Gelatine Capsule (CAL2K-4-GC). Creating a pellet of your sample will work in most cases to reduce the "Rate of Burn", but not always. In some instances you want to reduce the burning surface area of your sample, much like a fuse. The Gelatine Capsule created a fuse-like environment for your sample, because only one end of the capsule will ignite, causing your sample to burn evenly and slowly. Using a Gelatine Capsule is considered Spiking.
This is lite powder type sample. It is both lite as a mass and also prone to become airborne with the slightest breeze. Just by filling the bomb calorimeter with oxygen will disperse the sample out of the crucible. The problem arises that when the sample makes contact with the inside bomb walls, it will no longer combust. In this instance you can either make a pellet our of the sample, depending on the bonding properties of the sample or use a Gelatine Capsule. A Gelatine Capsule tends to be more reliable as it will also regulate the burning speed of the sample as these types of samples tend to have a high burning speed.
A volatile sample is typically a fuel that evaporates before your eyes. The problem is in the weight of the sample as it will reduce from the time of weighing to the time of firing. This is easily solved using a cover disk. The cover disk is a clear vinyl round disk, the same size as the top rim of the crucible. The cover disk (CAL2K-FLDS) is considered a spike as it will contribute to the energy of the sample. Stick the cover disk to the top of the crucible and then TARE the balance. Now inject the fuel sample through a small hole into the crucible. This technique works well for all volatile samples, even for fuel samples such as diesel.
SMALL VOLUME AND LOW CV
In some instances a small sample quantity with a weight less than 0.1g is only available. Our DDS Calorimeters are able to burn a sample with a mass of less than 0.1g, however it becomes a challenge to fill the bottom of the crucible in such a way do that the firing cotton makes contact with the sample consistently. A Gelatine Capsule will allow the sample to appear bigger in the crucible and provide a better firing repeatability. A low calorific value or CV can similarly be solved by using a Gelatine Capsule. Our DDS Calorimeters range is able to burn a low CV of around 1MJ and lower, but the problem comes from the low heat amount generated by the burning sample. To increase the amount of heat generated by the sample either increase the sample weight, but not more than what the crucible can accommodate, or spike it with a gelatine capsule. A Gelatine Capsule will create more heat for the bomb to register with a relatively small sample weight.
As a conclusion, the gelatine capsules are the most versatile and cost effective spiking aid available. Benzoic acid tablets are a close second. Both of which are provided in the start-up/installation kits provided with all our DDS Calorimeters.
CALORIMETER SPIKE MASS
This is the mass of the substance to be used for spiking, i.e. the gelatine capsule or benzoic acid tablet. The procedure to follow is to place the crucible on the balance and press TARE. Then add the substance to be used for spiking to the crucible. This mass value is known as the "Spike Mass" and must be entered manually into the calorimeter. The sample is then added. This mass is the spike item's mass plus the sample's mass and must be entered in as the total mass. This can be read directly from the balance if a balance is connected to the calorimeter.
STEP BY STEP - HOW TO ENTER THE SPIKE VALUE
- Place the empty crucible on the balance.
- Press "Tare" to clear the weight of the crucible and get a 0.0000g reading.
- Place one Benzoic Acid Tablet and take note of the new mass.
- Enter the mass value into the calorimeter as a "spike mass".
- Add the sample to the crucible and take note of the total mass.
- Enter the total mass into the calorimeter as "sample mass".
- Using the Up/Down Arrow (Navigation Keys) on the keyboard, go to the next item "Spike Value C".
- Press "=" and enter the energy value of the Benzoic Acid Tablet that you are using. This can be found on each bottle of DDS Certified Benzoic Acid bottle of tablets, e.g. 26.454 (MJ/g). This value needs to be entered each time you do a determination.
- Continue through the menu items, until you get to "Enter MASS, SID".
- Now place the prepared crucible (benzoic acid tablet with sample) in the lid assembly of the bomb vessel, close the vessel and fill it with the required 3000Kpa of oxygen.
- Place the vessel inside the calorimeter and close the calorimeter's lid.
- The calorimeter will now perform some standard checks and if all is okay, it will begin with the "Initial Phase", then "Fire", and finally the "Main" phase and then indicate that the determination has been completed.
Once the determination is complete, the result will display the calculated calorific value of the sample. The calorimeter will automatically take the end result, and subtract the spike energy and present you with the final result of your sample.
WHY IS BENZOIC ACID USED IN CALORIMETRY?
Benzoic Acid is a colourless crystalline solid and a simple carboxylic acid. The name is derived from gum benzoin, which was for a long time its only known source. Benzoic Acid occurs naturally in many plants and it serves as an intermediate in the biosynthesis of many secondary metabolites.
Although benzoic acid has many uses outside of bomb calorimetry, we will touch on the subject of its use in calorimetry. Benzoic acid is relatively non-toxic and occurs naturally in plants. Appreciable amounts of it has been found in most berries (around 0.05%) and is formed in apples after infection with a specific fungus.
Due to its constant value it is the ideal substance for calibration in bomb calorimeters. For this purpose we press the powdery substance into tablets of approximately 0.5 grams and 1.0 grams, with a calorific value of approximately 26.454KJ/g.Read More
CAL3K CONSUMABLES AND ACCESSORIES
Crucible Cover Discs (CAL2K-FLDS)
DDS supplies Crucible Cover discs in packs of 10 sheets. These cover discs are used to cover the crucible during the firing of combustible substances like fuel and oil and can be used with any of our combustion calorimeter systems.
Gelatine Capsules (CAL2K-4-GC)
DDS supplies gelatine capsules in packs of 100 capsules. The gelatine capsule hols the sample in the crucible during firing to prevent plattering i.e. powder samples, sawdust etc. and can be used with any of our combustion calorimeter systems.
Certified Benzoic Acid Tablets (CAL2K-4-BA)
DDS supplies benzoic acid tablets in bottles of 50 x 0.5g tablets. 2 bottles are included with our installation kits when purchasing a complete system and can be used with any of our combustion calorimeter systems.
For more information on any of the above consumables visit our consumables page or contact your nearest agent for a free quote.CAL3K ConsumablesFind your nearest agent